 Review
 Open access
 Published:
Compressed sensing MRI: a review from signal processing perspective
BMC Biomedical Engineering volume 1, Article number: 8 (2019)
Abstract
Magnetic resonance imaging (MRI) is an inherently slow imaging modality, since it acquires multidimensional kspace data through 1D free induction decay or echo signals. This often limits the use of MRI, especially for high resolution or dynamic imaging. Accordingly, many investigators has developed various acceleration techniques to allow fast MR imaging. For the last two decades, one of the most important breakthroughs in this direction is the introduction of compressed sensing (CS) that allows accurate reconstruction from sparsely sampled kspace data. The recent FDA approval of compressed sensing products for clinical scans clearly reflect the maturity of this technology. Therefore, this paper reviews the basic idea of CS and how this technology have been evolved for various MR imaging problems.
Background
Magnetic resonance imaging (MRI) exploits the nuclear magnetic resonance (NMR) phenomena to enable high contrast imaging. Since the first observation of NMR absorption in a molecular beam in 1938, the main research thrust was to understand and utilize the NMR phenomena for spectroscopy applications. Then, Lauterbur introduced in 1973 the gradient field to encode the spatial origin of the radio waves emitted from the nuclei of the object. This breakthrough allowed for multi dimensional imaging by NMR physics.
The idea of Lauterbur can be easily understood using kspace interpretation that describes MR sampling as Fourier encoding in 2D or 3D spaces. Specifically, data collected by an MRI scanner are samples of the spatial Fourier transform of an object image. Hence, in order to obtain an image without aliasing artifacts, kspace samples need to satisfy the Nyquist sampling criterion.
Despite this close link to signal sampling theory, until the first demonstration of the sensitivity encoding (SENSE) technique by Prussemann [1], MR imaging was not considered as an important research topic for signal processing. Specifically, Prussemann et al. [1] showed that spatial diversity information from coil sensitivity maps have additional information that can be exploited for fast signal acquisition. Furthermore, Sodickson et al. [2] proposed the simultaneous acquisition of spatial harmonics (SMASH). These works gave a birth of parallel imaging and iterative reconstruction methods, and has resulted in a flurry of novel ideas and algorithms, including Generalized Autocalibrating Partially Parallel Acquisitions (GRAPPA) by Griswold [3] and kt space method for cardiac imaging such as [4,5,6,7,8,9].
The common theme in these approaches is that the data redundancy can be exploited to reduce the required sampling rate. Because redundant data can be compactly represented in some transform domains, it is also closely related to the concept of “sparsity”. Originally investigated by Bresler and his students for on thefly Fourier imaging [10, 11], the sparsity regularization has become the main workhorse in modern accelerated MRI researches thanks to the introduction of the compressed sensing theory [12, 13]. Ever since the first demonstration of compressed sensing MRI by Lustig et al. [14, 15], the compressed sensing MRI has become the essential tools in modern MR imaging researches. In this paper, we will review these ideas in more detail.
Main text
MR forward models
Before we introduce the concept of compressed sensing MR, we begin with the discussion of MR forward model. Mathematically, the forward model for kspace measurement can be described by.
where r ∈ R^d and k ∈ R^{d}, d = 2, 3 denote the image domain and kspace coordinates, respectively, C is the number of the coil and the ith coil image γ^{i} is given by
Here, ν(r) denotes the contrast weighted bulk magnetization distribution, and s^{i}(r) is the corresponding coil sensitivity of the ith coil. Although the expression may give an impression that the measurement is two or three dimensional, this is indeed originated from a 1D measurement since the kspace trajectory is a function of time, i.e. k:= k(t), and we acquire one kspace sample at each time point t. This makes the MR imaging inherently slow, since we need to scan through a 3D object via 1D trajectories.
Dynamic MRI is another important MR technique to monitor dynamic processes such as brain hemodynamics and cardiac motion. Among the various forms of dy namic MR modeling, here we mainly focus on the kt formulation. Specifically, consider a discrete imaging equation for cartesian trajectory for simplicity. Because the samples along the readout direction are fully sampled, most of the dynamic MR formulation is applied separably after taking the Fourier transform along the read out direction. More specifically, let γ(s, t) denote the unknown image content (for example, proton density, T1/T2 weighted image, etc.) on the spatial coordinate s along the phase encoding line at time instance t. Then, the kt space measurement b(k, t) at time t is given by
where γ(s, t) is the spatiotemporal image which may be weighted by coil sensitivity map for the case of parallel imaging.
Throughout the paper, for simplicity we often use the operator notation for (1):
where [S^{i}] is a diagonal operator comprised of the ith coil sensitivity. Similarly, we use operator notation for (3).
or
for the case of parallel imaging.
Compressed sensing theory
Performance guarantees
Compressed sensing (CS) theory [12, 16, 17] addresses the accurate recovery of unknown sparse signals from underdetermined linear measurements and has become one of the main research topics in the signal processing area for the last two decades [18,19,20,21,22,23]. Most of the compressed sensing theories have been developed to address the socalled single measurement vector (SMV) problems [12, 16, 17]. More specifically, let m and n be positive integers such that m < n. Then, the SMV compressive sensing problem is given by
where y ∈ R^m, A ∈ R^m\×n, x ∈ R^n, and ϵ denotes the noise level. (P0) implies that the solution favours the sparsest solution.
Since (P0) requires a computationally expensive combinatorial optimization, greedy methods [24], reweighted norm algorithms [25, 26], or convex relaxation using the l_{1} norm [12, 27] have been widely investigated as alternatives. In particular, the convex relaxation approach addresses the following l_{1} minimization problem:
One of the important theoretical tools of CS is the socalled restricted isometry property (RIP), which enables us to guarantee the robust recovery of certain input signals [17]. More specifically, a sensing matrix \( \mathrm{A}\in {\mathrm{R}}^{m\times \mathrm{n}} \) is said to have a k restricted isometry property (RIP) if there is a constant 0 ≤ δ_{k} < 1 such that
for all x ∈ R^n with x_{0} ≤ k. It has been demonstrated that δ_{2k} < \( \sqrt{2}1 \) is sufficient for l_{1}/l_{0} equivalence [12]. For many classes of random matrices, the RIP condition is satisfied with high probability if the sampling pattern is incoherent and the number of measurements satisfies m ≥ ck log(n/k) for some constant c > 0 [17].
Optimization approaches
In practice, the following unconstrained form of optimization problem is often used:
where y is noisy measurement, λ is the regularization parameter, and Ψ refers to an analysis transform such that Ψx becomes sparse. Note that for the special case of Ψ = I, (8) is reduced to (P1).
One of the technical issues in solving (8) is that the cost function is not smooth at Ψx = 0, which makes the corresponding gradient illdefined. This leads to the development of various techniques, which are culminated as the new type of convex optimizaton theory called proximal optimization [28]. For example, the popular op timization methods such as forwardbackward splitting (FBS) [29], split Bregman iteration [30], alternating directional method of multiplier (ADMM) [31], Douglas Rachford splitting algorithm [32], and primaldual algorithm [33] have been devel oped to solve compressed sensing problems. However, the comprehensive coverages of these techniques deserves another review paper, so in this section we mainly review the ADMM algorithms which have been extensively used for compressed sensing MRI ever since its first introduction to compressed sensing MRI [34].
More specifically, we convert the problem (8) to the following constraint problem:
subject to
Then, the associated ADMM is given by
Now, each step of ADMM has a closedform solution. Algorithm 1 summarises the resulting ADMM iteration, where shrink_{1} denotes the softtresholding:
for the threshold value τ > 0.
In addition to the analysis prior in (8), the total variation (TV) penalty has been also extensively used for imaging applications, because the finite difference operator
can sparsify smooth images. Specifically, the TV minimisation problem assumes the following form:
where T V (x) is given by
where ‖∙‖_{1, p}, p = 1, 2 deotes the l_{1}/l_{p}mixed norm. In ddimensional space (e.g. d = 2 for images), the discretized implementation can be defined as
where ∇x(i) ∈ R^{d} denotes the gradient of x at the ith coordinate and n denotes the number of discretizated samples.
In order to apply ADMM for (12), we need to focus on the primal formulation of the total variation penalty:
Now, we define a splitting variable u(i) = ∇x(i) ∈ R^{d}. Then, the constraint opti mization formulation is given by
Then, the associated ADMM is given by
Each step has the closed form expression. Algorithm 2 summarises the TVADMM algorithm, where shrink_{vec,2}(x) for x ∈ Rd denotes the vector shrinkage [34]:
for the threshold value τ > 0.
Basic MR ingredients for compressed sensing
In order to apply compressed sensing theory for specific imaging applications, the unknown signal should be sparse in some transform domain, and the sensing matrix should be sufficiently incoherent. This section shows why these conditions can be readily satisfied in MR imaging, which is one of the main reasons that allows for successful applications of CS theory to MR imaging.
Sparsity of MR images
An MR image is rarely sparse in its own. However, one of the important observations that led to the successful applications of CS to MR is that the sparsity is closely related to signal redundancies. This is because redundant signals can be easily converted to sparse signals using some transforms.
Basically, there are three major directions that have been investigated in com pressed sensing MRI: 1) the spatial domain redundancy, 2) the temporal domain redundancy, and 3) the coil domain redundancy. For example, as shown in Fig. 1(a), natural images can be sparsely represented in finite difference or wavelet transform domain, although the image is not sparse in its own. This observation is the main idea that allows for total variation (TV) and wavelet transform approaches for image denoising, and reconstruction. Accordingly, TV and wavelets have been the main transforms that have been extensively used in most of the CS MRI researches. On the other hand, dynamic MR images such as cardiac cine, functional MRI, and MR parameter mapping have significant redundancy along the temporal dimension as shown in Fig. 1(b). For example, if the image is perfectly periodic, then temporal Fourier transform may be the optimal transform to sparsify the signal. However, in many dynamic MR problems, the temporal variations are dependent on the MR physics as well as specific motion of organs, so the analytic transform such as Fourier transform may not be an optimal solution, but more datadriven approaches such as PCA or dictionary learning are better options. Indeed, these observation leads to dictionary learning approaches that will be discussed later.
On the other hand, the coil redundancy is somewhat distinct compared to the spatial and temporal domain redundancy. As shown in Fig. 1(c), the redundancy of the coil image stems from the underlying common images, which results in cross channel redundancies:
where s^{i} denotes the ith coil sensitivity map and γ^{i} is the coil image. The rela tionship in (18) is easy to show since the coil image is given by (2). The main idea of the classical parallel MRI is to exploit this relationship. Specifically, the sensi tivity encoding (SENSE) [1] exploits the image domain redundancy described in (18), whereas the kspace domain approaches such as GRAPPA [3] exploits its dual relationship in the kspace:
where ∗ denotes the convolution, and \( {\widehat{s}}^j \), \( {\widehat{\gamma}}^j \)enote the Fourier transform of s^{j}and γ^{j}respectively. Later we will describe how these two expressions of the coil dimensional redundancies have been exploited in compressed sensing MRI.
Incoherent sampling pattern
In compressed sensing MRI, downsampling pattern is very important to impose the incoherence, but its realization is limited by MR physics. For example, in the 2D acquisition, the readout direction should be fully sampled, so there is only freedom along the phase encoding direction in designing incoherence sampling patterns. Figure 2(a)(c) shows the examples of realizable sampling patterns that have been exploited in the literature: a) cartesian undersampling, b) radial trajectory, and c) spiral trajectory. In particular, the radial and spiral trajectories, which have been studied even before the advanced of the compressed sensing [35], have more incoherent radial sampling patterns compared to the cartesian undersampling.
On the other hand, if we deal with 3D imaging or dynamic imaging, there are more rooms for sampling pattern design, since there are two dimensional degree of freedom. Figure 2(d) shows an example of 2D random sampling pattern. For the case of wavelet transformbased sparsity imposing prior, Lustig et al. [14] showed that the incoherence can be optimized by generalizing the notion of PSF to Transform Point Spread Function (TPSF). Specifically, TPSF measures how a single transform coefficient of the underlying object ends up influencing other transform coefficients of the measured undersampled object. To calculate the TPSF, a single point in the wavelet transform space at the ith location is transformed to the image space and then to the Fourier space. Once the corresponding Fourier space data is subsampled by the given downsampling pattern, its influence in the wavelet transform domain can be calculated by taking inverse transform followed by the inverse wavelet trans form. To have the best incoherence property, the side of TPSF should be as small as possible, such that the side lobe contribution can be removed by shrinkage op eration. This was used as the main criterion for sampling pattern design.
Historical milestones
The earliest application of the compressed sensing for MRI was done by Lustig and his collaborators [14]. The application of CS to dynamic MRI was pioneered by Jung et al. [18] and Gamper et al. [36], which was later significantly improved by Jung et al. [20]. The positive results of these earlier works have resulted in flurry of new ideas in static and dynamic MRI.
In terms of combining CS with parallel imaging, the first application for static imaging was done by Block et al. [37] by combining total variation penalty and parallel imaging; and by Liang et al. [38] in more general form using SENSE. In dynamic imaging, the works by Jung et al. [18] and Feng et al. [39] are among the first that combine the SENSE type parallel imaging with the kt domain compressed sensing. Combination with CS and parallel imaging using GRAPPA type constraints was pioneered by Lustig et al. [40]. Coil compression techniques have been also investigated to reduce the number of coils for CS reconstruction [41]. Feng et al. [42] later combined compressed sensing, parallel imaging, and goldenangle radial sampling for fast and flexible dynamic volumetric MRI, which has been approved for clinical use.
Aside from the standard l_{1} and TV penalty, several innovative sparsity inducing penalty have been used for compressed sensing MRI. For example, Trzasko et al. [43, 44] proposed a direct l_{0} mimimization approaches, whereas Knoll et al. [45, 46] proposed a generalized total variation approaches, and Sung et al. [47] combined the approximate message passing algorithm with parallel imaging. This idea was then extended to the dictionary learning [20, 48, 49] and motion compensation [20, 50]. Then, the idea of lowrank regularization was soon introduced [49, 51,52,53] thanks to the theoretical advances by Candes et al. [54]. The lowrank idea was further ex tended to structured lowrank approach for parallel imaging [55] and single coil imaging with the finite support [56]. The duality between the compressed sensing and lowrank Hankel matrix approaches were discovered by Jin and his colleagues [57,58,59,60] and Ongie et al. [61]. In particular, the unified framework for compressed sensing and parallel MRI in terms of lowrank Hankel matrix approaches was presented by Jin et al. [57] and its theoretical performance guarantees was also given in [62].
In the following, we provide more detailed reviews of these historical milestones in compressed sensing MRI.
Basic formulation of compressed sensing MRI
Although some MR images such as angiograms are already sparse in the pixel representation, more complicated images are rarely sparse, but only have a sparse representation in some transform domain, for example, in terms of spatial finite differences or their wavelet coefficients. Based on this observation, Lustig et al. [14] proposed the first compressed sensing MRI using spatial domain wavelet transform as a sparsifying transform. More specifically, the problem is formulated as
where ‖∙‖_{1} denotes the l_{1} norm, F is the Fourier transform, γ denotes the 2D complex image, Ψ is a either finite difference or spatial wavelet transform, D is the downsampling pattern, and y is the downsampled kspace measurement. Eq. (20) was solved using a nonlinear conjugate gradient method.
The resulting optimization algorithms are, however, computational expensive. For cartesian sampling trajectory, this problem can be overcome as follows. Specifically, the image update step for (20) in ADMM implementation can be summarized as
which is computationally expensive due to the matrix inverse to obtain γ. Instead of using a direct matrix inverse, by multiplying the Fourier transform F to both sides, we have
where \( \widehat{\gamma}=F\upgamma \). Note that a diagonal matrix D*D consisting of ones and zeros. The ones are at those diagonal entries that corresponds to the sampled locations in the kspace. Let Ω denote the sampled location. Then,
where P_{Ω} and P_{Ω}c denotes the projection on the sampling location Ω and its com plement Ω^{c}, respectively. Then, the image update γ^{(k + 1)}can be simply done by taking the fast Fourier transform (FFT).
The first experimental demonstration by Lustig et al. [14] clearly confirmed the efficiency of the compressed sensing algorithm, which has led to many other CS approaches using various sparsifying transform, and optimization algorithms, etc. For example, to deal with several hyperparameters to trade off between sparsity and data fidelity terms. Many research efforts have been made to quantity such trade off, provide different forms of reconstruction, even to propose hyperparameter free reconstruction method [63,64,65,66,67,68,69,70].
Advanced formulation for compressed sensing MRI
NonCartesian compressed sensing MRI
Compressed sensing with non Cartesian sampling has been also extensively stud ied, since they are really a great combination given the sampling behavior of non Cartesian sampling schemes and the incoherence requirement in compressed sens ing reconstruction. Aside from the radial and spiral sampling patterns discussed before, the works in [71,72,73,74,75,76,77,78,79,80,81,82,83] have fo cused on designing better sampling trajectories with good incoherent properties. For example, Haldar et al. [82] and Puy et al. [83] proposed random phase encod ing scheme to maximize the incoherency of the sensing matrix. However, one of the main technical issues associated with noncartesian compressed sensing MRI is that the fast reconstruction trick shown in (21) cannot be used, which increases the overall computational time.
Combination of parallel imaging with CS
Recall that parallel MRI (pMRI) [1, 3] exploits the diversity in the receiver coil sen sitivity maps that are multiplied by an unknown image. This provides additional spatial information for the unknown image, resulting in accelerated MR data acquisition through kspace sample reduction. Because the aim of the parallel imaging and CS approaches is similar, extensive research efforts have been made to syner gistically combine the two for further acceleration [18, 20, 38, 40, 84].
One of the most simplest approaches can be a SENSE type approach that explic itly utilizes the estimated coil maps to obtain an augmented compressed sensing problem [18, 20, 38, 84]. More specifically, if the coil sensitivity is known and given by the sensitivity[S^{i}], i = 1, ⋯, C,then the SENSE type compressed sensing MRI problem can be formulated as
The optimization framework is the standard optimization framework under sparsity constraint, so proximal optimization algorithms can be used to solve this problem.
On the other hand, l_{1}SPIRiT (l_{1}iTerative Selfconsistent Parallel Imaging Re construction) [40] utilizes the GRAPPA type constraint as an additional constraint for a compressed sensing problem:
where M is an image domain GRAPPA operator. In both approaches, an accurate estimation of coil sensitivity maps or GRAPPA kernel is essential to fully exploit the coil sensitivity diversity. To address this problem, Uecker et al. [85] developed a novel eigen space method to extract the coil sensitivity maps directly from the kspace data, which is one of the most popular methods widely used by MR researchers.
Blind compressed sensing MR using dictionary learning
Blind compressed sensing approaches attempted to simultaneously reconstruct the underlying image as well as the sparsifying transform from highly undersampled measurements. Ravishankar and his colleagues pioneered two distinct approaches  synthesis dictionary learning [48] and analysis transform learning [86]  when the underlying sparsifying transform is unknown a priori.
Synthesis dictionarybased BCS
More specifically, let P_{j}, j = 1 …, N represents the operator that extracts a m dimensional patch as a vector P_{j}x ∈ C^{m} from the image x, where N denotes the number of patches. Then, dictionary learning is to find the unknown dictionary \( D\in {\mathrm{R}}^{m\times \mathrm{Q}} \) and the corresponding sparse coefficient matrix \( C\in {\mathrm{C}}^{m\times \mathrm{n}} \) such that Y = DC, where Y = [P_{1}x P_{2}x…, P_{N}x]
The synthesis model allows each patch P_{j}x to be approximated by a linear combina tion Dc_{j} of a small number of columns from a dictionary \( D\in {\mathrm{C}}^{n\times \mathrm{K}} \), where c_{j} ∈ C^{K} is sparse. The columns of the learnt dictionary (represented by d_{k}, 1 ≤ k ≤ K) in (P0) are additionally constrained to be of unit norm in order to avoid the scaling ambiguity. The dictionary, and the image patch, are assumed to be much smaller than the image. This model can be used as a signal model, and Ravishankar et al. [48] proposed the following patchbased dictionary learning regularizer:
where C_{j} and d_{j} denotes the jth column of C and D, respectively. Then, the associated BCS formulation is given by
where A: = DF denotes the downsampled Fourier transform.
To address the optimization problem (P0), Ravishankar et al. [48] employed the following twostep alternating minimization algorithm. First, the following mini mization problem is solved by fixing x:
The KSVD algorithm [87] was used to learn the dictionary. For a given dictionary
D, the image update can be done by
These steps are alternated until convergence. The dictionary learning MRI have shown superior image reconstructions for MRI, as compared to nonadaptive compressed sensing schemes.
Sparsifying transformbased BCS
However, the BCS Problem (P0) is both nonconvex and NPhard. Approximate iterative algorithms for (P0) typically solve the synthesis sparse coding problem re peatedly, which makes them computationally expensive. In order to overcome some of the aforementioned drawbacks of synthesis dictionarybased BCS, Ravishankar et al. [86] proposed to use the sparsifying transform model in this work. Sparsify ing transform learning has been shown to be effective and efficient in applications, while also enjoying good convergence guarantees. Specifically, they used the follow ing transform learning regularizer:
where W ∈ C^{m\× m} denotes the unknown transform, the function Q(W) is a regualizer for the transform given by
The − log detW penalty eliminates degenerate solutions such as those with repeated rows. The ‖W‖_{F}penalty helps remove a scale ambiguity in the solution.
Then, with additional constraint ‖x‖ ≤ E, the overall optimization problem is given by
where A = DF again denotes the downsampled Fourier transform. One of the important advantages of (P1) is that there exists a closedform update for W, so the computationally expensive dictionary learning step can be avoided.
Kt methods for compressed sensing dynamic MRI
Dynamic MRI is a technique to acquire temporally varying MR sequences such as cardiac cine, perfusion, timeresolved angiography, functional MRI, etc. In dynamic MRI, there exists significant redundancies along the temporal directions, which can be extensively studied in various compressed sensing approaches.
kt SPARSE
The kt SPARSE by Lustig et al. [88] is an earliest version of compressed sensing dynamic MRI. Recall that the kspace measurement b(k, t) at time t is given by
Another application of the Fourier transform along the temporal direction
results in the following 2D Fourier relationship:
where ρ(s, f) denotes the temporal Fourier transform of γ(s, t).
Note that p(s, f) is usually sparse because the periodic motions from heart or slow varying motions from fMRI can be easily sparsified using the temporal Fourier transform. Accordingly, the kt data can be represented as mapping from the spatial temporal image:
where A: = DF and D is a kt downsampling pattern, and F now becomes a 2D Fourier transform. In addition to the temporal Fourier transform, the authors in [88] used the wavelet transform in the spatial dimension to exploit the spatial redundancy. Then, kt SPARSE is formulated based on the following optimization problem:
where W denotes the spatial wavelet transform, respectively.
Kt FOCUSS
Preliminary CS dynamic MRI approaches [36, 88] were seemingly different from the classical kt approach such as kt BLAST/SENSE [5]. One of the most impor tant contributions of the kt FOCUSS by Jung et al. [18, 20] was to reveal that the compressed sensing dynamic MRI is not very different from the classical kt approaches, but rather it can be obtained by a very simple modification of the classical kt BLAST/SENSE to ensure significant performance improvement.
More specifically, rather than using wavelet transform, kt FOCUSS exploited the xf domain sparsity. Then, a standard compressed sensing formulation would be:
to enforce the sparseness in xf image ρ. However, there exists two additional novel ties in the kt FOCUSS. First, rather than directly enforcing the sparseness of the xf image, the kt FOCUSS further sparsifies the xf image using the initial estimate.
Specifically, let ρ_{0} be the predictable initial estimate of ρ. Then, the residual
should be much more sparse than the original spatiotemporal image. Second, rather than directly using the l_{1} minimization, kt FOCUSS employed the reweighted norm approaches. This results in the following equivalent minimization problem:
Then, the normal equations with respect to x and v are given by
where V is a diagonal matrix whose ith diagonal element is v_{i}. This result in the following FOCUSS iteration:
where the weighting matrix is given by
One of the most powerful observations in [18, 20] was that the first iteration of (38) has very similar form to the classical kt BLAST/SENSE algorithm, except the power factor of weighting matrix. This observation led to an innovative idea to con vert kt BLAST/SENSE to compressed sensing approach. More specifically, by using incoherence sampling patterns, multiple iterations and correct weighting factor for the diagonal matrix, the authors of kt FOCUSS [18, 20] clearly demonstrated the performance improvement. This observation suggested that the improvement by the classical kt algorithms such as kt BLAST/SENSE [5] was not from the Bayesian perspective as the original authors of [5] had claimed, but indeed is originated from exploiting the sparsity int the spatiotemporal domain [18, 20]. Furthermore, by simply modifying the weight factor and sampling patterns, several additional iter ation can significantly improve the performance of kt BLAST/SENSE.
Another powerful aspect of kt FOCUSS was that the idea can be easily extended to exploit the sparsity in other transform domains. For example, the residual step in (36) can be interpreted as sparsity promoting step by subtracting the temporal mean images. Thus, Jung et al. [20] proposes motion estimated and compensated modification of kt FOCUSS to make the residual signal much sparser. More specifi cally, rather than subtracting the temporal mean values, they subtracted the motion estimated frame. Note that motion estimation and compensation (ME/MC) is an essential step in video coding that uses motion vectors to exploit the temporal re dundancies between frames [89, 90]. In order to employ ME/MC within dynamic MRI, there are several technical issues to address. First, at least one reference frame is required. This issue can be easily resolved if we acquire fully sampled data in one frame as often done in dynamic MRI. The main technical difficulty, however, comes from the existence of the low quality current frame. Fortunately, this issue can be addressed using an additional reconstruction step before the ME/MC [20].
In addition, the spatiotemporal signals can be further sparsified using datadriven transform:
where D denotes the learned temporal dictionary based from the images, whereas c denotes the coefficients. Then, the imaging problem can be formulated as
For example, in order to find the temporal basis that can sparsify ρ, Jung et al. [18, 20] performed the the singular value decomposition (SVD) after the image reconstruction, then the new dictionary is used to estimate the new coefficients. This procedure is closely related to the partial separable function (PSF) [91] and kt SLR (kt sparse and lowrank decomposition) [49], which will be reviewed soon.
Partially separable function (PSF) approach
In the PSF model by Liang et al. [91], the kt samples are assumed to be decomposed in the following form
for some data dependent spectral and temporal basis function \( {\left\{{\Psi}_l(k)\right\}}_{l=1}^L \) and \( {\left\{{\varnothing}_l(t)\right\}}_{l=1}^L \) . Thanks to the partially separable assumption, the socalled Casorati Matrix B for the fully sampled kt data given by
has at most rank L.
In dynamic CS MRI, many of the kt samples are missing and we are interested in finding the missing components. Hence, by utilizing the lowrankness of the Casorati matrix, the missing kt samples can be estimated using a low rank matrix completion algorithm. In particular, the authors in [92, 93] proposed the following matrix factorization approach:
where A denotes the kt sampling operator that indicates the missing samples by 0, and \( U\in {\mathrm{C}}^{m\times \mathrm{L}} \), \( V\in {\mathrm{C}}^{m\times \mathrm{L}} \) is used for the low rank matrix factorization B = U V ^{H}, and the optimization is performed for U and V alternatingly by fixing the other matrix using the previous estimate. Because the exact rank of B is not known, the authors proposed the incremented powerFactorization (IRFP) algorithm [92], where (40) starts with L = 1 by increasing order with the initialization of the power factorization of L + 1 from that of L. To avoid an overfitting, the algorithm is terminated as soon as the data fidelity is below some threshold values.
Kt SLR: Kt sparse and low rank approach
kt Sparse and Low Rank Approach (kt SLR) model by Lingala et al. [49] is a more systematic way of learning both basis and sparse coefficients. In this approach, the spatiotemporal signal ρ(x, t) is first rearranged in a matrix form
Then, using the low rank prior, the optimisation problem can be formulated as
\( \underset{\Gamma}{\min }{\left\Vert bA\left(\Gamma \right)\right\Vert}^2+\uplambda \upvarphi \left(\Gamma \right) \),
where A: = DF is now a downsampled Fourier transform. Here, the rank prior is approximated using the general class of Schatten pfunctionals, specified by
where {σ_{i}} denotes the singular values of Γ.
In dynamic imaging applications, the images in the time series may have sparse wavelet coefficients or sparse gradients. In addition, if the intensity profiles of the voxels are periodic (e.g., cardiac cine), they may be sparse in the Fourier domain. Based on this observation, Lingala et al. [49] proposed additional sparsity inducing penalty in specified basis sets along with the lowrank property to further improve the recovery rate. Specifically, they chose the 2D wavelet transform to sparsify each of the images in the time series, while can be a 1D Fourier transform to exploit the pseudoperiodic nature of motion. Then, the resulting composite minimization problem can be formulated as
Kspace structured lowrank approaches
Basic theory
Compared to the standard compressed sensing approaches, kspace structured low rank approaches such as SAKE [55], LORAKS [56], ALOHA [57, 59, 60] and GIRAF [61] are relatively new, but has significant potentials in MRI imaging. These ap proaches are all derived by the kspace convolution relationship and can be used for both static and dynamic imaging. So we first discuss a matrix representation of the convolution. For simplicity, we will consider the 1D notation.
Specifically, consider a fully sampled kspace measurement from the multichannel coils:
where γ^{i} denotes the unknown ith coil images and y^{i} corresponds to its kspace data. Note that the matrix representation of a kspace convolution of y^{i} with a dtap filter h is given by
where \( \mathcal{C}\left({y}^i\right) \) denotes the convolution matrix constructed by the vector y^{i}:
and h¯ denotes a vector that reverses the order of the elements. If we extract the n − drows of the convolution matrix with n − d > d, we can obtain the following Hankel structured matrix:
By defining Y = [y^{1}, ⋯, y^{C}], we can further defined the extended Hankel matrix
In the following, we will explain how these Hankel structured matrix has been utilised for accelerated MRI.
SAKE
A calibrationless parallel imaging reconstruction method, termed simultaneous au tocalibrating and kspace estimation (SAKE), is a datadriven, coilbycoil recon struction method that does not require a separate calibration step for estimating coil sensitivity information [55]. SAKE is based on the following observation in GRAPPA:
which implies that the ith kspace measurement can be represented as the linear combination of the filtered kspace data from other coils. In matrix form, this recursive relationship implies the existence of the null space of the Hankel matrix.
In other word, H_{dC}(Y) is lowranked. Therefore, SAKE solves the following low rank matrix completion problem to interpolate the missing kspace data:
where P_{Ω}(·) denotes the projection on the measured kspace samples on the index set Ω. The problem was solved using the iterative singular value shrinkage method [94].
LORAKS
The lowrank modeling of localspace neighborhoods (LORAKS) [56] was inspired by the finite support condition. More specifically, if the object γ has finite support, we can easily find the function w such that
This results in a convolution relationship in kspace
where y and h denote the Fourier spectrum of γ and w, respectively. Therefore, this gives a single channel version of the lowrank condition, which results in the following rank minimization problem:
ALOHA and GIRAF
Annihilating filterbased low rank Hankel matrix (ALOHA) approach [57,58,59,60, 62] and GIRAF (Generic Iterative Reweighted Annihilating Filter) [61] can be considered as the full generalization of SAKE and LORAKS for general class of signals with the finite rate of innovations (FRI) for MR measurements. Moreover, the approaches have unified the parallel imaging and compressed sensing as a k space interpolation with performance guarantees [62], and can be used for artifact correction [58]. This section describes the fundamental dual relationship between transform domain sparsity and low rankness in reciprocal domain, which is the key ingredient. For better readability, we provide here a high level description by assuming 1D signals.
The Fourier CS problem of our interest is to recover the unknown signal x(t) from the Fourier measurement:
In classical Nyquist sampling, to avoid aliasing artefacts, the grid size should be at most:
when the support of the time domain signal x(t) is τ. Then, discrete Fourier data at the Nyquist rate is defined by:
We also define a length (r + 1)annihilating filter hˆ[k] for xˆ[k] that satisfies
The existence of the minimum length finite length annihilating filter has been ex tensively studied for FRI signals [95,96,97]. Let r + 1 denotes the minimum size of annihilating filters that annihilates discrete Fourier data \( \widehat{\mathrm{x}}\left[\mathrm{k}\right] \). Then, a dtap anni hilating filter h with d > r + 1 can be easily obtained by convolving an appropriate size FIR filter with the minimum length annihilating filter. In matrix form, this is equivalent to
where the Hankel structure matrix \( {\mathcal{H}}_{\mathrm{d}}\left(\hat{x}\right) \) is constructed as
Assume that min(n − d + 1, d) > r. Then, we can show the following low rank property [62]:
Thanks to the lowrankness of the associated Hankel matrix, the missing kspace data can be easily interpolated using the following lowrank matrix completion [62]:
where P_{Ω} is the projection operator on the sampling location Ω. Moreover, as shown in [62], the lowrank matrix completion approach (55) does not compromise any optimality compared to the standard Fourier CS.
Note that signals may not be sparse in the image domain, but can be sparsified in a transform domain. In fact, this was the main idea of the compressed sensing. Specifically, the signal x of our interest is a nonuniform spline that can be represented by:
where L denotes a constant coefficient linear differential equation that is often called the continuous domain whitening operator in [98, 99]:
and w is a continuous sparse innovation:
For example, if the underlying signal is piecewise constant, we can set L as the first differentiation. In this case, x corresponds to the total variation signal model. Then,
by taking the Fourier transform of (56), we have
where
Accordingly, the Hankel matrix \( \mathcal{H}\left(\hat{z}\right) \) from the weighted spectrum zˆ(ω) satisfies the following rank condition:
Thanks to the lowrankness, the missing Fourier data can be interpolated using the following matrix completion problem:
or, for noisy Fourier measurements \( \widehat{\mathrm{y}} \),
where ⊙ denotes the Hadamard product, \( \widehat{\mathrm{l}} \) and \( \widehat{\mathrm{x}} \) denotes the vectors composed of full samples of \( \widehat{\mathrm{l}} \)(ω) and \( \widehat{\mathrm{x}} \)(ω), respectively. After solving (P_{w}), the missing spec tral data xˆ(ω) can be obtained by dividing by the weight, i.e. \( \widehat{x}\left(\omega \right)=\mathrm{m}\left(\upomega \right)/\widehat{l}\left(\omega \right) \) assuming that \( \widehat{l}\left(\omega \right)\ne 0 \).
The idea can be generalized for any transform domain sparse signals as long as the transform can be represented using shiftinvariant filters. Wavelet domain sparse signal belongs to this class. In this case, the weight kernel in the Fourier domain is obtained as the spectrum of the subband filters [57,58,59,60, 62]. For example, Fig. 3 showed the construction of the Hankel matrix for the MR parameter mapping, where the kspace weighting from wavelet weighting is applied only along the phase encoding direction, whereas no weighing is applied along the tdomain since the correspond temporal spectrum is already sparse [59].
The idea can be also easily generalised to the parallel imaging by exploiting (19). Specifically, (19) can be equivalently represented using the matrix representation:
This implies that an exten^ded Hankel matrix \( {\mathcal{H}}_{\mathrm{d}\mid \mathrm{C}}\left(\left[{\widehat{\gamma}}^1\cdots {\widehat{\gamma}}^C\right]\right) \) in (46) is low ranked.
For example, the multichannel construction of Hankel matrix for MR parameter mapping [59] is shown in Fig. 4.
One of the most important advantages of the Hankel matrix formulation is that the coil diversity can be readily exploited in addition to the image domain redundancy in a unified framework. This makes the separate coil sensitivity estimation unnecessary.
Clinical applications
Since the compressed sensing MRI allows significant acceleration of MR acquisition, it has been extensively applied for various clinical applications such as fast cardiac MRI, whole heart MRI, dynamic contrast enhanced (DCE)MRI, diffusion MRI, spectroscopic, etc., that usually require significant acquisition time using standard methods.
For example, Otazo et al. [84] applied the CS method to the first pass cardiac perfusion MRI and demonstrated feasibility of 8fold acceleration in vivo imaging using standard coil arrays. They showed that CS method results in similar temporal fidelity and image quality to GRAPPA with 2fold acceleration [84]. Hsiao et al. [100] applied combined parallel imaging and compressed sensing to achieve 4D phase contrast for the quantification of cardiac flow and ventricular volumes pediatric patients during congenital heart MRI examinations. Vincent et al. [101] employed CS to evaluate LV function and volumes and found that CS strategy with the single breath hold provided similar results to multi breathhold imaging protocols.
For freebreathing contrastenhanced multiphase liver MRI, Chandarana et al. [102] showed that a combination of compressed sensing, parallel imaging, and radial kspace sampling demonstrated the feasibility of breathhold cartesian T1 weighted imaging. Espagnet et al. [103] employed goldenangle radial sparse parallel technique for DCEMRI to evaluate the permeability characteristics of the pituitary gland.
For diffusion MRI, Landman et al. [104] showed that CS reconstruction using standard data can resolve crossing fibers similar to a standard qball approach using much richer data with longer acquisition time. Kuhnt et al. [105] showed that High Angular Resolution Diffusion Imaging (HARDI) + CS is a promising approach for fiber tractography in clinical practice.
Finally, for the spectroscopic imaging, Larson et al. [106] developed a CS method for acquiring hyperpolarized 13C data using multiband excitation pulses and achieved 2 s temporal resolution with full volumetric coverage of a mouse. Geethanath et al. [107] demonstrated a potential reduction in acquisition time by up to 80% or more for hydrogen 1 MR spectroscopic imaging using CS, with negligible loss of clinical information.
With the commercially available CS reconstruction methods, we expect to see more clinical applications of CS in the near future.
Conclusions
Nowdays, compressed sensing has become an mature technology, as reflected by recent approval by FDA. Major vendors have started to sell the compressed sensing reconstruction softwares, and many clinical researchers have been evaluating its clinical usefulness.
Despite of this maturity, some of the main technical issues of the compressed sensing are 1) the computational complexity of the algorithm is relative high, and at high acceleration, image quality degradation is still reported. Although the recent stateofthe art CS techniques such as structured Hankel matrix approach can address the quality degradation problems, it also increases the computational complexity, which may interfere the clinical workflow.
Fortunately, for the last two years, the MR image reconstruction field have been rapidly changed thanks to the successful demonstration of of the deep learning based MR reconstruction technologies [108,109,110,111,112,113]. The sudden popularity of deep learning approaches can be attributed to the realtime reconstruction in spite of the significant improvement of the image quality. Thus, when originally presented, these techniques were regarded as totally different technology that is nothing to do with the compressed sensing. However, recent theoretical analysis [114] showed that the deep convolutional neural network is closely related to the Hankel matrix decomposition. Therefore, we can still argue that the compressed sensing MRI has renewed interests in the form of deep learning, and it will be interesting to see how this exciting and rapidly evolving field will develop for the coming decades.
Abbreviations
 ALOHA:

Annihliating filterbased lowrank Hankel matrix approach
 CS:

Compressed sensing
 FOCUSS:

FOCal Underdetermined System Solver
 GRAPPA:

Generalized Autocalibrating Partially Parallel Acquisitions
 l1SPiRIT:

l1iTerative Selfconsistent Parallel Imaging Reconstruction
 LORAKS:

Lowrank modeling of localspace neighborhoods
 MRI:

Magnetic resonance imaging
 SAKE:

Simultaneous autocalibrating and kspace estimation
 SENSE:

Sensitivity encoding
References
Pruessmann KP, Weigher M, Scheidegger MB, Boesiger P. SENSE: Sensitivity encoding for fast MRI. Magn. Reson. Med. 1999;42(5):952–62.
Sodickson DK, Manning WJ. Simultaneous acquisition of spatial harmonics (SMASH): fast imaging with radiofrequency coil arrays. Magn Reson Med. 1997;38(4):591–603.
Griswold MA, Jakob PM, Heidemann RM, Nittka M, Jellus V, Wang J, Kiefer B, Haase A. Generalized autocalibrating partially parallel acquisitions(GRAPPA). Magn. Reson. Med. 2002;47(6):1202–10.
Madore B, Glover GH, Pelc NJ. Unaliasing by Fourierencoding the overlaps using the temporal dimension(UNFOLD), applied to cardiac imaging and fMRI. Magn. Reson. Med. 1999;42(5):813–28.
Tsao J, Boesiger P, Pruessmann KP. kt BLAST and kt SENSE: Dynamic MRI with high frame rate exploiting spatiotemporal correlations. Magn Reson Med. 2003;50(5):1031–42.
Kozerke S, Tsao J, Razavi R, Boesiger P. Accelerating cardiac cine 3D imaging using kt BLAST. Magn. Reson. Med. 2004;52:19–26.
Sharif B, Derbyshire JA, Faranesh AZ, Bresler Y. Patientadaptive reconstruction and acquisition in dynamic imaging with sensitivity encoding (paradise). Magn Reson Med. 2010;64(2):501–13.
Tsao J, Kozerke S, Boesiger P, Pruessmann KP. Optimizing spatiotemporal sampling for kt BLAST and kt SENSE: Application to highresolution realtime cardiac steadystate free precession. Magn. Reson. Med. 2005;53:1372–82.
Hansen MS, Kozerke S, Pruessman KP, Boesiger P, Pedersen EM, Tsao J. One the influence of training data quality in kt BLAST reconstruction. Magn. Reson. Med. 2004;52:1175–83.
Bresler, Y., Gastpar, M., Venkataramani, R.: Image compression onthefly by universal sampling in fourier imaging systems. In: Proc. 1999 IEEE Information Theory Workshop on Detection, Estimation, Classification, and Imaging, pp. 48 (1999)
Ye, J.C., Bresler, Y., Moulin, P.: A selfrefencing levelset method for image reconstruction from sparse Fourier samples. In: to Appear in Proc. IEEE Workshop in Variational and Level Set Methods in Computer Vision, Vancouver, Canada (2001)
Candes E, Romberg J, Tao T. Robust uncertainty principles: Exact signal reconstruction from highly incomplete frequency information. IEEE Trans. Inf. Theory. 2006;52(2):489–509.
Donoho DL. Compressed sensing. IEEE Trans. Inf. Theory. 2006;52(4):1289–306.
Lustig M, Donoho D, Pauly JM. Sparse MRI: The application of compressed sensing for rapid MR imaging. Magn Reson Med. 2007;58(6):1182–95.
Lustig M, Donoho DL, Santos JM, Pauly JM. Compressed sensing MRI. IEEE Signal Process Mag. 2008;25(2):72–82.
Candes EJ, Romberg JK, Tao T. Stable signal recovery from incomplete and inaccurate measurements. Commun Pure Appl Math. 2006;59(8):1207–23.
Candes E, Tao T. Decoding by linear programming. IEEE Trans. Info. Theory. 2005;51(12):4203–15.
Jung H, Ye JC, Kim EY. Improved k–t BLAST and k–t SENSE using FOCUSS. Phys. Med. Biol. 2007;52:3201–26.
Ye JC, Tak S, Han Y, Park HW. Projection reconstruction MR imaging using FOCUSS. Magn Reson Med. 2007;57(4):764–75.
Jung H, Sung K, Nayak KS, Kim EY, Ye JC. kt FOCUSS: a general compressed sensing framework for high resolution dynamic MRI. Magn Reson Med. 2009;61(1):103–16.
Chen GH, Tang J, Leng S. Prior image constrained compressed sensing (PICCS): a method to accurately reconstruct dynamic CT images from highly undersampled projection data sets. Med. Phys. 2008;35:660.
Cotter SF, Rao BD. Sparse channel estimation via matching pursuit with application to equalization. IEEE Trans. Commun. 2002;50:374–7.
Wagadarikar A, John R, Willett R, Brady D. Single disperser design for coded aperture snapshot spectral imaging. Applied Optics. 2008;47(10):44–51.
Tropp JA. Just Relax: convex programming methods for identifying sparse signals in noise. IEEE Trans. Inf Theory. 2006;52(3):1030–51.
Gorodnitsky IF, Rao BD. Sparse signal reconstruction from limited data using FOCUSS: A reweighted minimum norm algorithm. IEEE Trans Signal Process. 1997;45(3):600–16.
Candes EJ, Wakin MB, Boyd SP. Enhancing sparsity by reweighted l1 minimization. J Fourier Anal Appl. 2008;14(5):877–905.
Chen SS, Donoho DL, Saunders MA. Atomic decomposition by basis pursuit. SIAM J Sci Comput. 1999;20(1):33–61.
Bauschke HH, Combettes PL. Convex analysis and monotone operator theory in Hilbert spaces. Vol. 408. New York: Springer; 2011.
Combettes PL, Wajs VR, et al. Signal recovery by proximal forwardbackward splitting. Multiscale Model Simul. 2006;4(4):1168–200.
Goldstein T, Osher S. The split Bregman method for L1 regularized problems. SIAM J Imag Sci. 2009;2(2):323–43.
Wang Y, Yang J, Yin W, Zhang Y. A new alternating minimization algorithm for total variation image reconstruction. SIAM J Imag Sci. 2008;1(3):248–72.
Combettes PL, Pesquet JC. A Douglas–Rachford splitting approach to nonsmooth convex variational signal recovery. IEEE J. Sel. Top. Sign. Proces. 2007;1(4):564–74.
Chambolle A, Pock T. A firstorder primaldual algorithm for convex problems with applications to imaging. J. Math. Imaging Vision. 2011;40(1):120–45.
Ramani S, Fessler JA. Parallel MR image reconstruction using augmented Lagrangian methods. IEEE Trans. Med. Imaging. 2011;30(3):694–706.
Meyer CH, Hu BS, Nishimura DG, Macovski A. Fast spiral coronary artery imaging. Magn Reson Med. 1992;28(2):202–13.
Gamper U, Boesiger P, Kozerke S. Compressed sensing in dynamic MRI. Magn Reson Med. 2008;59(2):365–73.
Block KT, Uecker M, Frahm J. Undersampled radial MRI with multiple coils. Iterative image reconstruction using a total variation constraint. Magn Reson Med. 2007;57(6):1086–98.
Liang D, Liu B, Wang J, Ying L. Accelerating SENSE using compressed sensing. Magn. Reson. Med. 2009;62(6):1574–84.
Feng L, Srichai MB, Lim RP, Harrison A, King W, Adluru G, Dibella EV, Sodickson DK, Otazo R, Kim D. Highly accelerated realtime cardiac cine MRI using k–t SPARSESENSE. Magn Reson Med. 2013;70(1):64–74.
Lustig M, Pauly JM. SPIRiT: Iterative selfconsistent parallel imaging reconstruction from arbitrary kspace. Magn Reson Med. 2010;64(2):457–71.
Zhang T, Pauly JM, Vasanawala SS, Lustig M. Coil compression for accelerated imaging with Cartesian sampling. Magn Reson Med. 2013;69(2):571–82.
Feng L, Grimm R, Block KT, Chandarana H, Kim S, Xu J, Axel L, Sodickson DK, Otazo R. Goldenangle radial sparse parallel MRI: combination of compressed sensing, parallel imaging, and goldenangle radial sampling for fast and flexible dynamic volumetric MRI. Magn Reson Med. 2014;72(3):707–17.
Trzasko J, Manduca A. Fast spiral coronary artery imaging. IEEE Trans. Med. Imaging. 2009;28(1):106–21.
Trzasko JD, Haider CR, Borisch EA, Campeau NG, Glockner JF, Riederer SJ, Manduca A. SparseCAPR: highly accelerated 4D CEMRA with parallel imaging and nonconvex compressive sensing. Magn Reson Med. 2011;66(4):1019–32.
Knoll F, Bredies K, Pock T, Stollberger R. Second order total generalized variation (TGV) for MRI. Magn Reson Med. 2011;65(2):480–91.
Knoll F, Clason C, Bredies K, Uecker M, Stollberger R. Parallel imaging with nonlinear reconstruction using variational penalties. Magn Reson Med. 2012;67(1):34–41.
Sung K, Daniel BL, Hargreaves BA. Location constrained approximate message passing for compressed sensing MRI. Magn Reson Med. 2013;70(2):370–81.
Ravishankar S, Bresler Y. MR image reconstruction from highly undersampled kspace data by dictionary learning. IEEE Trans. Med. Imaging. 2011;30(5):1028–41.
Lingala SG, Hu Y, DiBella E, Jacob M. Accelerated dynamic MRI exploiting sparsity and lowrank structure: kt SLR. IEEE Trans. Med. Imaging. 2011;30(5):1042–54.
Asif MS, Hamilton L, Brummer M, Romberg J. Motionadaptive spatiotemporal regularization for accelerated dynamic MRI. Magn Reson Med. 2012;7:800–12.
Ak Cakaya M, Basha TA, Goddu B, Goepfert LA, Kissinger KV, Tarokh V, Manning WJ, Nezafat R. Lowdimensionalstructure selflearning and thresholding: Regularization beyond compressed sensing for MRI Reconstruction. Magn Reson Med. 2011;66(3):756–67.
Trzasko JD, Manduca A. Calibrationless parallel MRI using CLEAR. In: IEEE Conference Record of the Forty Fifth Asilomar Conference on Signals, Systems and Computers (ASILOMAR); 2011. p. 75–9.
Yoon H, Kim KS, Kim D, Bresler Y, Ye JC. Motion adaptive patchbased lowrank approach for compressed sensing cardiac cine MRI. IEEE Trans. Med. Imaging. 2014;33(11):2069–85.
Cand’es EJ, Recht B. Exact matrix completion via convex optimization. Found. Comput. Math. 2009;9(6):717–72.
Shin PJ, Larson PE, Ohliger MA, Elad M, Pauly JM, Vigneron DB, Lustig M. Calibrationless parallel imaging reconstruction based on structured lowrank matrix completion. Magn Reson Med. 2014;72(4):959–70.
Haldar JP. Lowrank modeling of localspace neighborhoods (LORAKS) for constrained MRI. IEEE Trans. Med. Imaging. 2014;33(3):668–81.
Jin KH, Lee D, Ye JC. A general framework for compressed sensing and parallel MRI using annihilating filter based lowrank Hankel matrix. IEEE Trans Computational Imag. 2016;2(4):480–95.
Jin KH, Um JY, Lee D, Lee J, Park SH, Ye JC. MRI artifact correction using sparse+ lowrank decomposition of annihilating filterbased Hankel matrix. Magnetic Reson Med. 2017;78(1):327–40.
Lee D, Jin KH, Kim EY, Park SH, Ye JC. Acceleration of MR parameter mapping using annihilating filterbased low rank hankel matrix (ALOHA). Magn Reson Med. 2016;76(6):1848–68.
Lee J, Jin KH, Ye JC. Referencefree singlepass EPI Nyquist ghost correction using annihilating filterbased low rank Hankel matrix (ALOHA). Magn Reson Med. 2016;76(8):1775–89.
Ongie G, Jacob M. Offthegrid recovery of piecewise constant images from few fourier samples. SIAM J Imaging Sci. 2016;9(3):1004–41.
Ye JC, Kim JM, Jin KH, Lee K. Compressive sampling using annihilating filterbased lowrank interpolation. IEEE Trans Inf Theory. 2017;63(2):777–801.
Bauer S, Markl M, Honal M, Jung B. The effect of reconstruction and acquisition parameters for GRAPPAbased parallel imaging on the image quality. Magn Reson Med. 2011;66(2):402–9.
Huang F, Lin W, Duensing GR, Reykowski A. A hybrid method for more efficient channelbychannel reconstruction with many channels. Magn Reson Med. 2012;67(3):835–43.
Trzasko JD, Bao Z, Manduca A, McGee KP, Bernstein MA. Sparsity and lowcontrast object detectability. Magn Reson Med. 2012;67(4):1022–32.
Johnson KM, Block WF, Reeder SB, Samsonov A. Improved least squares MR image reconstruction using estimates of kspace data consistency. Magn Reson Med. 2012;67(6):1600–8.
Park S, Park J. Adaptive selfcalibrating iterative GRAPPA reconstruction. Magn Reson Med. 2012;67(6):1721–9.
Liang D, DiBella EV, Chen RR, Ying L. kt ISD: dynamic cardiac MR imaging using compressed sensing with iterative support detection. Magn Reson Med. 2012;68(1):41–53.
Chang Y, Liang D, Ying L. Nonlinear GRAPPA: A kernel approach to parallel MRI reconstruction. Magn Reson Med. 2012;68(3):730–40.
Huang F, Lin W, Duensing GR, Reykowski A. kt sparse GROWL: Sequential combination of partially parallel imaging and compressed sensing in kt space using flexible virtual coil. Magn Reson Med. 2012;68(3):772–82.
Jung B, Stalder AF, Bauer S, Markl M. On the undersampling strategies to accelerate timeresolved 3D imaging using ktGRAPPA. Magn Reson Med. 2011;66(4):966–75.
Piccini D, Littmann A, NiellesVallespin S, Zenge MO. Spiral phyllotaxis: the natural way to construct a 3D radial trajectory in MRI. Magn Reson Med. 2011;66(4):1049–56.
Lin W, B¨ornert P, Huang F, Duensing GR, Reykowski A. Generalized GRAPPA operators for wider spiral bands: Rapid selfcalibrated parallel reconstruction for variable density spiral MRI. Magn Reson Med. 2011;66(4):1067–78.
Pipe JG, Zwart NR, Aboussouan EA, Robison RK, Devaraj A, Johnson KO. A new design and rationale for 3d orthogonally oversampled kspace trajectories. Magn Reson Med. 2011;66(5):1303–11.
Pang Y, Vigneron DB, Zhang X. Parallel travelingwave MRI: A feasibility study. Magn Reson Med. 2012;67(4):965–78.
Wang H, Liang D, King KF, Nagarsekar G, Chang Y, Ying L. Improving GRAPPA using crosssampled autocalibration data. Magn Reson Med. 2012;67(4):1042–53.
Witschey WR, Cocosco CA, Gallichan D, Schultz G, Weber H, Welz A, Hennig J, Zaitsev M. Localization by nonlinear phase preparation and kspace trajectory design. Magn Reson Med. 2012;67(6):1620–32.
Addy NO, Wu HH, Nishimura DG. Simple method for MR gradient system characterization and kspace trajectory estimation. Magn Reson Med. 2012;68(1):120–9.
Zahneisen, B., Hugger, T., Lee, K.J., LeVan, P., Reisert, M., Lee, H.L., Assl¨ander, J., Zaitsev, M., Hennig, J.: Single shot concentric shells trajectories for ultra fast fMRI. Magn Reson Med 68(2), 484–494 (2012)
Turley DC, Pipe JG. Distributed spirals: a new class of threedimensional kspace trajectories. Magn Reson Med. 2013;70(2):413–9.
Layton KJ, Gallichan D, Testud F, Cocosco CA, Welz AM, Barmet C, Pruessmann KP, Hennig J, Zaitsev M. Single shot trajectory design for regionspecific imaging using linear and nonlinear magnetic encoding fields. Magn Reson Med. 2013;70(3):684–96.
Haldar JP, Hernando D, Liang ZP. Compressedsensing MRI with random encoding. IEEE Trans Med Imag. 2011;30(4):893–903.
Puy G, Marques JP, Gruetter R, Thiran JP, Ville DVD, Vandergheynst P, Wiaux Y. Spread spectrum magnetic resonance imaging. IEEE Trans Med Imag. 2012;31(3):586–98.
Otazo R, Kim D, Axel L, Sodickson DK. Combination of compressed sensing and parallel imaging for highly accelerated firstpass cardiac perfusion MRI. Magn Reson Med. 2010;64(3):767–76.
Uecker M, Lai P, Murphy MJ, Virtue P, Elad M, Pauly JM, Vasanawala SS, Lustig M. ESPIRiT–an eigenvalue approach to autocalibrating parallel MRI: where SENSE meets GRAPPA. Magn Reson Med. 2014;71(3):990–1001.
Ravishankar S, Bresler Y. Efficient blind compressed sensing using sparsifying transforms with convergence guarantees and application to magnetic resonance imaging. SIAM J Imag Sci. 2015;8(4):2519–57.
Aharon M, Elad M, Bruckstein A. KSVD: An algorithm for designing overcomplete dictionaries for sparse representation. IEEE Trans Signal Process. 2006;54(11):4311–22.
Lustig M, Santos JM, Donoho DL, Pauly JM. kt SPARSE: High frame rate dynamic MRI exploiting spatiotemporal sparsity. In: Proceedings of the 13th Annual Meeting of ISMRM, Seattle, vol. 2420; 2006.
Le Gall D. MPEG: A video compression standard for multimedia applications. Commun ACM. 1991;34(4):46–58.
Jung H, Ye JC. Motion estimated and compensated compressed sensing dynamic magnetic resonance imaging: What we can learn from video compression techniques. Int J Imag Syst Technol. 2010;20(2):81–98.
Haldar, J.P., Liang, Z.P.: Spatiotemporal imaging with partially separable functions: a matrix recovery approach. In: IEEE International Symposium on Biomedical Imaging, pp. 716–719 (2010). IEEE
Haldar JP, Hernando D. Rankconstrained solutions to linear matrix equations using powerfactorization. IEEE Signal Process Letters. 2009;16(7):584–7.
Zhao B, Haldar JP, Christodoulou AG, Liang ZP. Image reconstruction from highly undersampledspace data with joint partial separability and sparsity constraints. IEEE Trans Med Imag. 2012;31(9):1809–20.
Cai JF, Cand’es EJ, Shen Z. A singular value thresholding algorithm for matrix completion. SIAM J Optimization. 2010;20(4):1956–82.
Vetterli M, Marziliano P, Blu T. Sampling signals with finite rate of innovation. IEEE Trans Signal Process. 2002;50(6):1417–28.
Dragotti PL, Vetterli M, Blu T. Sampling moments and reconstructing signals of finite rate of innovation: Shannon meets Strang–Fix. IEEE Trans Signal Process. 2007;55(5):1741–57.
Maravic I, Vetterli M. Sampling and reconstruction of signals with finite rate of innovation in the presence of noise. IEEE Trans Signal Process. 2005;53(8):2788–805.
Unser M, Tafti PD, Sun Q. A unified formulation of Gaussian versus sparse stochastic processes–Part I: Continuousdomain theory. IEEE Trans Inf Theory. 2014;60(3):1945–62.
Unser M, Tafti PD, Amini A, Kirshner H. A unified formulation of Gaussian versus sparse stochastic processes–Part II: Discretedomain theory. IEEE Trans Inf Theory. 2014;60(5):3036–51.
Hsiao A, Lustig M, Alley MT, Murphy M, Chan FP, Herfkens RJ, Vasanawala SS. Rapid pediatric cardiac assessment of flow and ventricular volume with compressed sensing parallel imaging volumetric cine phasecontrast MRI. Am J Roentgenol. 2012;198(3):250–9.
Vincenti G, Monney P, Chaptinel J, Rutz T, Coppo S, Zenge MO, Schmidt M, Nadar MS, Piccini D, Ch’evre P, et al. Compressed sensing single–breathhold CMR for fast quantification of LV function, volumes, and mass. JACC: Cardiovasc Imag. 2014;7(9):882–92.
Chandarana H, Feng L, Block TK, Rosenkrantz AB, Lim RP, Babb JS, Sodickson DK, Otazo R. Freebreathing contrastenhanced multiphase MRI of the liver using a combination of compressed sensing, parallel imaging, and goldenangle radial sampling. Investigative Radiol. 2013;48(1):10–16.
Espagnet MR, Bangiyev L, Haber M, Block K, Babb J, Ruggiero V, Boada F, Gonen O, Fatterpekar G. Highresolution DCEMRI of the pituitary gland using radial kspace acquisition with compressed sensing reconstruction. Am J Neuroradiol. 2015;36(8):1444–9.
Landman BA, Bogovic JA, Wan H, ElShahaby FEZ, Bazin PL, Prince JL. Resolution of crossing fibers with constrained compressed sensing using diffusion tensor MRI. NeuroImage. 2012;59(3):2175–86.
Kuhnt D, Bauer MH, Egger J, Richter M, Kapur T, Sommer J, Merhof D, Nimsky C. Fiber tractography based on diffusion tensor imaging compared with highangularresolution diffusion imaging with compressed sensing: initial experience. Neurosurgery. 2013;72(0 1):165.
Larson PE, Hu S, Lustig M, Kerr AB, Nelson SJ, Kurhanewicz J, Pauly JM, Vigneron DB. Fast dynamic 3D MR spectroscopic imaging with compressed sensing and multiband excitation pulses for hyperpolarized 13C studies. Magn Reson Med. 2011;65(3):610–9.
Geethanath S, Baek HM, Ganji SK, Ding Y, Maher EA, Sims RD, Choi C, Lewis MA, Kodibagkar VD. Compressive sensing could accelerate 1H MR metabolic imaging in the clinic. Radiology. 2012;262(3):985–94.
Hammernik K, Klatzer T, Kobler E, Recht MP, Sodickson DK, Pock T, Knoll F. Learning a variational network for reconstruction of accelerated MRI data. Magn Reson Med. 2018;79(6):3055–71.
Han Y, Yoo J, Kim HH, Shin HJ, Sung K, Ye JC. Deep learning with domain adaptation for accelerated projectionreconstruction mr. Magn Reson Med. 2018;80(3):1189–205.
Zhu B, Liu JZ, Cauley SF, Rosen BR, Rosen MS. Image reconstruction by domaintransform manifold learning. Nature. 2018;555(7697):487.
Schlemper J, Caballero J, Hajnal JV, Price AN, Rueckert D. A deep cascade of convolutional neural networks for dynamic mr image reconstruction. IEEE Trans Med Imag. 2018;37(2):491–503.
Wang, S., Su, Z., Ying, L., Peng, X., Zhu, S., Liang, F., Feng, D., Liang, D.: Accelerating magnetic resonance imaging via deep learning. In: Biomedical Imaging (ISBI), 2016 IEEE 13th International Symposium On, pp. 514–517 (2016). IEEE
Kwon K, Kim D, Park H. A parallel MR imaging method using multilayer perceptron. Med Phys. 2017;44(12):6209–24.
Ye JC, Han Y, Cha E. Deep convolutional framelets: A general deep learning framework for inverse problems. SIAM J Imag Sci. 2018;11(2):991–1048.
Acknowledgements
The author would like to thank his students, Yoseob Han and Dongwook Lee, for providing figures. The author also like to thank two anonymous reviewers who comments have significantly improve the quality of the paper.
Funding
This work is supported by National Research Foundation of Korea, Grant number NRF2016R1A2B3008104.
Availability of data and materials
Data sharing not applicable to this article as no datasets were generated or analysed during the current study.
Author information
Authors and Affiliations
Contributions
The author has read and approved this article.
Corresponding author
Ethics declarations
Ethics approval
Not applicable.
Consent for publication
Not applicable
Competing interests
The author declares that he has no competing interests.
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Ye, J.C. Compressed sensing MRI: a review from signal processing perspective. BMC biomed eng 1, 8 (2019). https://doi.org/10.1186/s424900190006z
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s424900190006z